CAMBRIDGE ACCUSENSE ATM2400 DRIVER DETAILS:
|File Size:||11.1 MB|
|Supported systems:||ALL Windows 32x/64x|
|Price:||Free* (*Free Registration Required)|
CAMBRIDGE ACCUSENSE ATM2400 DRIVER
Map can be used to distinguish the variation at different points in the layout.
The scale on the right-top displays the cambridge accusense atm2400 range of readings. Also can select number of pages layout planes of that type. Gradient Gradient tab displays continuous gradient of the sensor-grid layout; and so is useful to understand the general behavior of the layout.
Continuous Maps The following is that for a 3X5 layout plane. The probes Airflow, Temperature etc are combined in a single page.
Specs & Details
Please refer Web Server functionality cambridge accusense atm2400 learn more. Discrete Maps To get a non-continuous map, according to custom number of ranges of values, do the following: 1 Display Options by selecting the Options button on top of the screen see figure below. The live-chart of the layout is useful to understand the fluctuation in values.
You can select different chart-types and can export the chart to different image formats. Also individual and group readings can be averaged for the user-specified period of time.
Cambridge ATM Multipoint Airflow & Air Temperature Measurement System for sale online eBay
Last 5 snapshots of readings with calculated averages will be continously shown below the live matrix. Calibration reporting enable the user to send the list of sensors that need calibration. If the user is not opted to hide or to remind laterhe is prompted to send the report when a sensor is about to be used for data collection. On starting the cambridge accusense atm2400 collection, if any of the sensors exceeded calibration date, the following screens will appear.
You can start a reporting from Help menu too, anytime when some sensors are connected. Welcome Screen Ensure the success of your testing: www. How to proceed? The following screen lists sensors that are out-of-calibration. NOTE: The last column has the reported date, if the sensor is already reported for calibration earlier. Send cambridge accusense atm2400 email.
Cambridge Accusense ATM 2400 Air Velocity Sensor (FS58506-R01) Capacitance Meter
Then you will get the next screen, where you can change the receiving email address etc. Troubleshooting: Cambridge accusense atm2400 you get an error like the one following, use another 'Mail Server'. The use of standard CAT-5 or better Ethernet cable allows the ATM hub to be located up to meters from the host computer before an active repeater is required.
Open the Network control panel Select Change adapter settings www. Select the proeprties of Local Area Connection d.
Select Use the following IP address button, and then enter the IP address found on cambridge accusense atm2400 bottom label of the ATM USB to Ethernet adapter housing, for the last three numbers, choose a value that is close to but not the same as the adapter module. For the Subnet mask, use Default IP is Please wait for the device to restart and connect with new IP. Select the IP address from the list box to connect to the device. Please wait for a while to start the data transfer www.
Are you using the latest software version? Did you try restarting the instrument, computer, and software?ATMe Airflow & Temperature Measurement System Full compatibility with any of our Cambridge Accusense USB Sensors (Air Velocity, Temperature. Cambridge AccuSense ATM Cambridge accusense atm2400 Device.
Vendor, Filename, Description, Operating system, Version, Filesize. _5aaee37bbdef3.